The Binding Site IMMPROVE™
serum paraprotein QA scheme (QA003) was the first scheme to include sFLC analysis. To date, there have been more than 70 distributions over 12 years, and over 300 laboratories worldwide currently participate in the scheme. Four serum samples are issued per year and results can be returned via the IMMPROVE website. Methodologies assessed include serum protein electrophoresis (SPE), serum immunofixation electrophoresis (sIFE) and sFLC assays (Chapter 4
). Laboratories are able to return quantitative results for IgG, IgA, IgM, β2
-microglobulin, κ sFLCs, λ sFLCs and the κ/λ sFLC ratio. A comprehensive report is returned to each laboratory in which sFLC results are categorised according to nephelometric/turbidimetric instrument type to allow user group comparisons to be made. The report also includes examples of electrophoresis gels and an interpretative comment regarding the sample and results.
Electrophoresis and sFLC results from a typical QA003 report (Distribution 129, August 2014) are shown in Figures 39.1
. Whilst 82% of laboratories reported that no monoclonal protein was visible by SPE or sIFE, the majority of participants reporting Freelite® results identified monoclonal κ FLCs of approximately 500 mg/L. A small sub-group of laboratories reported significantly lower κ sFLC concentrations. This sub-group primarily consisted of laboratories using another (non-Freelite) method for FLC measurement. This distribution illustrates that absolute FLC values reported by other FLC assays may be significantly different to those reported by Freelite (Chapter 8
In May 2013 the Binding Site launched a pilot quality assurance scheme for Hevylite analysis (QA003.H). Four samples are distributed each year and over 40 laboratories currently participate in the scheme. The scheme requires samples to be initially screened by SPE and sIFE, followed by Hevylite assays with the HLC pair selected as indicated by the immunofixation results. Participants receive a comprehensive report that includes analysis of qualitative results for SPE/sIFE and quantitative results for HLC assays that correspond with the monoclonal intact immunoglobulin type. Examples from a typical report are shown in Figures 39.4 and 39.5.